I am not sure if there is something wrong with my communications settings, or if I’m using the SCPI commands wrong. I tried using the FETC? In the 4th shot you see the error message when I try to read the status byte. Pre-register for updates on when this product will be available for ordering. Call us toll free Send It.
|Date Added:||25 January 2015|
|File Size:||43.63 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Our Member Store gives you access to the following advantages: The driver on the BK Precision site didn’t work the first time I tried it — it had the same timeout error — but seems to be working now.
DMMs are slow devices. We also accept wire transfers, cheques and purchase orders on approved credit. Be the first to know when we have news, discounts, special offers and promotions.
B&K Precision 2831E True RMS Bench Multimeter
Most Active Software Boards: The problem is reading value from FETC? This product includes one lens. In the second screen shot, you can see more of the communications settings. It looks like NI Package Manager is not installed; could this be the problem?
I have tried every single step from solution here, but i still can not read from BK E. Yes, Sign Me Up!
Have you tried out the instrument driver that is available here? Thank you very much!
We accept these payment methods: Request Quote on ITM. Pre-register for updates on when this product will be available for ordering.
Providing the finest test equipment solutions since 1992
Yes you should also change the timeout for the VISA session to some useful value. In the 4th shot you see the error message when I try to read the status byte.
So add an additional property to the VISA session where you set the baudrate and everything and set this to a value of or something. I also tried the above queries and writes with the “Enable termination character” box unchecked, but it didn’t seem to make a difference.
BK Precision E Dual Display 4 1/2 True RMS Bench Multimeter – at the Test Equipment Depot
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Submit Request Add More Items.
If I disable that command, the rest of the VI works without error. Message 3 of 6. Therefore, 2831w must select at least one lens with this item.
B&K Precision E 4 1/2 Digit True RMS Bench Multimeter | Transcat Canada
Save my login information. I am not sure if there is something wrong with my communications settings, or if I’m using the SCPI commands wrong. Precisioh also puts the meter into “remote” mode. You can unsubscribe at any time. The driver in there should be ready made and give you all you need, without having to bother about such details.
Message 6 of 6. Test Read and Write without flushing.